FOR SURGEONS (DURING CAPSULOTOMY)
Position the Zepto® suction cup horizontally on the capsule plane prior to initiating suction. Tilting upwards will prevent Zepto apposition with the capsule.
Always slide the push rod all the way back to the large portion of the neck to ensure as suction is initiated. This ensures full suction is delivered to create the Zepto capsulotomy edge.
Wait for the flow of bubbles to more or less stop before delivering energy for the capsulotomy. This ensures full apposition of the nitinol capsulotomy element to the capsule to create the strong capsulotomy edge.
During Zepto release, always wait for small amount of OVD/BSS to exit the corneal incision. This indicates that the Zepto suction cup has been floated off the capsule and can now be withdrawn from the eye.
Do not touch the Zepto tip (suction cup and capsulotomy ring) with fingers or an instrument prior to use in surgery.
Do not tilt the Zepto suction cup upwards as this will inhibit capsule apposition.
Do not leave the push rod in the neck as suction is initiated. This significantly impedes the delivery of suction.
Do not deliver capsulotomy energy before bubbles have stopped flowing. The capsule will not be optimally apposed to the nitinol capsulotomy element and an incomplete or weak capsulotomy may result.
Do not move Zepto off the capsule until it is fully released as this may damage the capsulotomy edge. Wait until the assistant has pushed the roller dispenser forward and a small amount of OVD/BSS has exited the corneal incision.